Results 1 - 3 of 3
Results 1 - 3 of 3. Search took: 0.017 seconds
|Sort by: date | relevance|
[en] In long range ultrasonic testing, a phased array probe composed of multiple identical transducers with an uniform interval of one quarter wavelength is usually used for the transmission or reception directivity control. This paper shows that the propagation directions of individual echoes can be identified in real time by displaying the inputs of a process for summing the constitution reception signals after compensating the phase difference due to the transducer interval, together with the output of the process. A constructive interference of the constitution echoes indicates a forward direction echo propagating along an intended direction while a destructive interference implies a reverse direction echo propagating along the direction opposite to the intended one
[en] Separate and layered transmitting and receiving spiral coil arrays had been used for the optimized individual applications of magnetostrictive type long-range ultrasonic testing. In this study, it was demonstrated that when taking advantages of the spiral coils with the leg width reduced to be half of the previous one and of the empty spaces due to the decrease of leg width, the transmitting and receiving coil array can be arranged to form a single layer structure allowing more easy use and fabrication. Because of the number of turns of the receiving coils that is proportional to their leg width, the sensitivity of the single layer coil array was about half that of the corresponding double layer coil array while it allowed the receiving amplifier to get faster recovery from a saturation due to the main bang echo. It was also found that the two types of coil arrays have almost the same performances in the view points of signal-to-noise ratio and directivity control.
[en] Like the phase array based ultrasonic system, complicated electronics has been used for real time signal processing in the magnetostrictive long-range ultrasonic testing(LRUT) system. This study shows that the software approach including the phase compensation, noise filtering and waveform transformation takes advantage rather than the previous hardware approach. Furthermore, it is possible for the software approach to be able more flexible and efficient realtime signal processing. These results will contribute to a cost-effective LRUT system and analysis of the inspection data.